Calibration of dissolution test apparatus (USP apparatus 1 and 2)
This calibration Standard Operating Procedure (SOP) describes all the individual steps necessary for calibrating dissolution test apparatus type 1 (basket apparatus) and type 2 (paddle apparatus) in accordance with USP requirements and cGMP (current good manufacturing practices). As for any calibration in the pharmaceutical environment, the calibration of dissolution test apparatus also needs to be described in a procedure and followed by the analyst in line with regulations in the US Pharmacopeia under General Chapters Dissolution.
This Standard Operating Procedure (SOP) details the individual steps involved in calibration with USP disintegrating tablets (Prednisone) and non-disintegrating tablets (salicylic acid). This includes, but is not limited to, calibration of the rotation and wobble of the dissolution spindle.
All necessary documentation requirements are listed in this Standard Operating Procedure (SOP) and its corresponding attachments. Both methods - Apparatus 1 (baskets) or Apparatus 2 (paddles) - and a combination of the two are described on 12 pages, including a 3-page protocol attached to the SOP. This SOP template is ready to use and needs only a small amount of site-specific modification.
Table of Content
1. Regulatory Basis, Reference Documents
4. Responsibilities and Accountabilities
4.1 Quality Control
4.2 Quality Assurance
5.1 General Requirements
5.3 Chemical Calibration
5.3.2 USP Dissolution Calibrator Disintegrating Type (Prednisone Tablets, 10 Mg) - Procedure
126.96.36.199 Prednisone Standard Stock Solution
188.8.131.52 Prednisone Standard Solution
184.108.40.206 Dissolution Calibration Using USP Prednisone Calibrator Tablets – Apparatus 1 (baskets)
220.127.116.11 Dissolution Calibration Using USP Prednisone Calibrator Tablets – Apparatus 2 (paddles)
5.3.3 USP Dissolution Calibrator Non-Disintegrating Type (Salicylic Acid Tablets, 300 mg) - Procedure
18.104.22.168 Chemical calibration
22.214.171.124 Dissolution Calibration Using USP Salicylic Acid Calibrator Tablets – Apparatus 1 (baskets)
126.96.36.199 Dissolution Calibration Using USP Salicylic Acid Calibrator Tablets – Apparatus 2 (paddles)
5.4 Physical/Mechanical Calibration
5.5 Routine Examination
5.6 Calibration Failure
7.1 Calibration Stickers
7.2 Chemical Calibration Protocol
7.3 Physical/Mechanical Calibration Protocol
|SKU||Calibration of dissolution test apparatus|
What you will get:
Delivery item: Word Document
You will be able to enter your data into the Word document.
Please note: We do not deliver hard copies.