HPLC Calibration - GMP SOP
This HPLC calibration Standard Operating Procedure (SOP) describes all the individual steps necessary for calibrating a HPLC in accordance with cGMP (current good manufacturing practices). As for any calibration in the pharmaceutical environment, the calibration of the HPLC also needs to be described in a procedure and followed by the analyst in line with regulations. To avoid getting cited e.g. by the FDA with a 483 for not having documentation regarding calibration for a HPLC in the Quality control laboratories or not having acceptance criteria or the interval of calibration set prior conduct.
This Standard Operating Procedure (SOP) details all the necessary individual steps involved in the HPLC calibration.
All necessary documentation requirements are listed on 22 pages Standard Operating Procedure (SOP) and its corresponding 8 attachments. This SOP template is ready to use and needs only a small amount of site-specific modification.
Table of Content
1. Regulatory Basis, Reference Documents
4.2 Calibration of the pumps
4.3 Gradient Accuracy Check
4.3.2 Data Evaluation
4.4 Calibration of Detector
4.4.1 Wavelength Accuracy Check
4.4.2 Data Analysis
4.4.3 Detector Intensity (Linearity) Test
4.5 Column oven temperature check
4.6 Auto sampler Injection Linearity
Annexure – I ® Flow Rate Accuracy Check
Annexure – II ® Gradient Accuracy check.
Annexure – III ® Wavelength Accuracy Check.
Annexure – IV ®Detector Intensity (Linearity) Check.
Annexure – V ®Column oven temperature check.
Annexure – VI ® Auto sampler injection linearity
|SKU||HPLC Calibration - GMP SOP|
What you will get:
Delivery item: Word Document
You will be able to enter your data into the Word document.
Please note: We do not deliver hard copies.
Here's how it works: Download your copy of this document and if for any reason your aren't completely satisfied after SEVEN ENTIRE DAYS just give us a call, and we'll personally guarantee you get a complete refund.